This article describes step-by-step how to set the ventilator when using the Adaptive Support Ventilation (ASV) mode on ARDS patients.
ASV targets the optimal respiratory rate for the low compliance of an ARDS patient and already targets low tidal volumes as recommended in the ARDS guidelines1.
For any given minute volume, ASV determines the optimal respiratory rate and tidal volume associated with the minimal work of breathing and minimal driving pressure. It applies lung-protective ventilation with low tidal volumes (VT), and limited Pplateau and driving pressure in passive ICU patients with different lung conditions. In ARDS patients, ASV applies individualized lung-protective ventilation. 2,3
Step 1: Determine patient height and select gender
Step 2: Initial settings
- %MinVol setting of 120% (this corresponds to 120 ml/kg IBW/min)
- PEEP setting of 8 cmH2O
- Oxygen 100% adjust according to your blood gas examinations
Step 3: Adjust %MinVol and Pasvlimit
Increase or decrease the %MinVol to reach the target PaCO2 in passive patients. In spontaneously breathing patients, increase or decrease the %MinVol to change the level of pressure support.
Be aware that tidal volumes delivered by ASV can be limited by means of the pressure limitation (Pasvlimit) setting. Adjust the Pasvlimit according your Pplateau measurement results if needed. Normally a Pasvlimit ≤30 cmH2O is sufficient; in severe cases with low chest wall compliance or high airway resistance, a higher Pasvlimit up to 35 cmH2O might be needed. Verify your PEEP and Pasvlimit settings according to transpulmonary pressure measurement. Keep the driving pressure below 14 cmH204 and the transpulmonary driving pressure below 10 cmH20.
Step 4: Consider an early recruitability assessment
An RCT5 investigating high PEEP and prolonged high pressure RMs showed harm to the lung, suggesting that the protocol used in this RCT should be avoided. In those trials however, patients were not screened for recruitability and they were exposed to high pressures over a long time. We recommend using the P/V Tool Pro for an early assessment of recruitability; if a patient is a responder, recruitment should be carried out promptly with the P/V Tool Pro.
For more information:
(Click here to dowload these documents in other languages.)
- JM Arnal, MS Saoli, D Novotni, A Garnero. Driving pressure automatically selected by INTLLiVENT-ASV in ICU patients. Intensive Care Med Exp 2016; 4 (Suppl 1):A602.
- Arnal, Jean-Michel, Mathieu Saoli, and Aude Garnero. "Airway and transpulmonary driving pressures and mechanical powers selected by INTELLiVENT-ASV in passive, mechanically ventilated ICU patients." Heart & Lung (2019).
- Amato, Marcelo BP, et al. "Driving pressure and survival in the acute respiratory distress syndrome." New England Journal of Medicine 372.8 (2015): 747-755.
- Writing Group for the Alveolar Recruitment for Acute Respiratory Distress Syndrome Trial Investigators, Cavalcanti AB, Suzumura EA, et al. Effect of Lung Recruitment and Titrated Positive End-Expiratory Pressure (PEEP) vs Low PEEP on Mortality in Patients With Acute Respiratory Distress Syndrome: A Randomized Clinical Trial. JAMA 2017;318:1335-45.
For additional information on COVID-19:
ESICM information: https://www.esicm.org/resources/coronavirus-public-health-emergency/
Current evidence about COVID-19: https://jamanetwork.com/journals/jama/pages/coronavirus-alert
Centers for Disease Control and Prevention, CDC: https://www.cdc.gov/coronavirus/2019-ncov/hcp/clinical-guidance-management-patients.html
- COVID-19: Respiratory treatment in critical care – current recommendations
- How to set INTELLiVENT®-ASV® for a (COVID-19) ARDS patient
The content of this Knowledge Base is intended for informational purposes only. Hamilton Medical AG provides no warranty with respect to the information contained in this Knowledge Base and reliance on any part of this information is solely at your own risk. For detailed instructions on operating your Hamilton Medical device, please refer to the official Hamilton Medical Operator’s Manual for the respective device.
The content of this Knowledge Base is intended for informational purposes only. Medin Medical AG provides no warranty with respect to the information contained in this Knowledge Base and reliance on any part of this information is solely at your own risk. For detailed instructions on operating your Medin Medical device, please refer to the official Medin Medical Operator’s Manual for the respective device.