What does this small bump at the beginning of expiration represent?
Answer: Premature cycling. The patient is still making an inspiratory effort when the ventilator cycles to expiration. The image below shows esophageal pressure, as well as pressure and flow, with the blue, vertical line indicating the end of the patient’s inspiratory effort. You should try to prolong insufflation by lowering the percentage of expiratory trigger sensitivity (ETS).
The content of this newsletter is for informational purposes only and is not intended to be a substitute for professional training or for standard treatment guidelines in your facility. Any recommendations made in this newsletter with respect to clinical practice or the use of specific products, technology or therapies represent the personal opinion of the author only, and may not be considered as official recommendations made by Hamilton Medical AG. Hamilton Medical AG provides no warranty with respect to the information contained in this newsletter and reliance on any part of this information is solely at your own risk.
The content of this Knowledge Base is intended for informational purposes only. Medin Medical AG provides no warranty with respect to the information contained in this Knowledge Base and reliance on any part of this information is solely at your own risk. For detailed instructions on operating your Medin Medical device, please refer to the official Medin Medical Operator’s Manual for the respective device.