What causes the distortions of pressure and flow during the second breath?
Answer: Reverse triggering. This refers to a diaphragmatic muscle contractions triggered by ventilator insufflations. The patient in this case was highly sedated and ventilated in volume-control mode. The second mechanical insufflation was triggered by the ventilator, inducing an inspiratory muscle contraction that can be seen from the decrease in esophageal pressure (vertical dotted line). This inspiratory effort was sufficient to trigger another breath, resulting in double inspiration with a large tidal volume.
Akoumianaki E, Lyazidi A, Rey N, Matamis D, Perez-Martinez N, Giraud R, Mancebo J, Brochard L, Richard JM. Mechanical ventilation-induced reverse-triggered breaths: a frequently unrecognized form of neuromechanical coupling. Chest 2013;143:927-938.
The content of this newsletter is for informational purposes only and is not intended to be a substitute for professional training or for standard treatment guidelines in your facility. Any recommendations made in this newsletter with respect to clinical practice or the use of specific products, technology or therapies represent the personal opinion of the author only, and may not be considered as official recommendations made by Hamilton Medical AG. Hamilton Medical AG provides no warranty with respect to the information contained in this newsletter and reliance on any part of this information is solely at your own risk.
The content of this Knowledge Base is intended for informational purposes only. Medin Medical AG provides no warranty with respect to the information contained in this Knowledge Base and reliance on any part of this information is solely at your own risk. For detailed instructions on operating your Medin Medical device, please refer to the official Medin Medical Operator’s Manual for the respective device.