Is this a) auto-triggering, b) an ineffective effort, c) early cycling, or d) double triggering?
The ventilator is triggered in the absence of an inspiratory effort by the patient.
How to recognize it:
Look for a delivered mechanical breath showing no drop in airway pressure (1) at the beginning of the inspiratory phase. This is because the patient does not make an active inspiratory effort. Did you know that over 60% of health care professionals surveyed found it difficult to recognize and treat asynchronies in adult and pediatric patients?1. See the list of possible causes and solutions below.
Common possible causes:
- Trigger threshold set too low
- Air leaks in the endotracheal tube cuff, ventilator circuit, or chest tube
- Flow oscillations (water or secretion in the circuit, cardiac oscillations)
Increase the inspiratory trigger setting in order to avoid oscillations (if not heart-associated) and leaks.
Reference: Result of internal survey conducted online from May 27 - June 23, 2019 with 155 voluntary participants worldwide (respiratory therapists, physicians, nurses, biomed/technician/engineer, paramedics, other).
The content of this newsletter is for informational purposes only and is not intended to be a substitute for professional training or for standard treatment guidelines in your facility. Any recommendations made in this newsletter with respect to clinical practice or the use of specific products, technology or therapies represent the personal opinion of the author only, and may not be considered as official recommendations made by Hamilton Medical AG. Hamilton Medical AG provides no warranty with respect to the information contained in this newsletter and reliance on any part of this information is solely at your own risk.
The content of this Knowledge Base is intended for informational purposes only. Medin Medical AG provides no warranty with respect to the information contained in this Knowledge Base and reliance on any part of this information is solely at your own risk. For detailed instructions on operating your Medin Medical device, please refer to the official Medin Medical Operator’s Manual for the respective device.