Heliox therapy with the HAMILTON-G5/S1
Heliox, a mixture of helium and oxygen, may be indicated for patients in cases of acute and life-threatening upper airway obstruction.
This action is taken as a temporary measure to provide a decrease in the patient’s work of breathing while the cause of the obstruction is treated. Administering heliox can make it easier to ventilate, because its lower density can allow a patient to produce inspiratory and expiratory flows with less turbulence.
The Heliox option available with the HAMILTON-G5/S1* ventilator can be activated in just a few quick steps, allowing you to administer heliox and treat the cause of the obstruction.
This video provides you with instructions on how to change the gas source and activate the Heliox option, including putting the device into standby mode first and calibrating the flow sensor after each change.
* HAMILTON-S1 not available in all markets.
The content of this Knowledge Base is intended for informational purposes only. Hamilton Medical AG provides no warranty with respect to the information contained in this Knowledge Base and reliance on any part of this information is solely at your own risk. For detailed instructions on operating your Hamilton Medical device, please refer to the official Hamilton Medical Operator’s Manual for the respective device.